Long Distance Signal Sensor Magnetrol Level Indicator
|(1) Center-center distance:||150~6000mm|
|(2) Medium density:||More than 0.45g/cm3|
|(3) Nominal pressure:||PN63~PN320(x 0.1MPa)|
|(4) Material:||SS304, 316L or customized|
|(5)Operating temperature:||-190 ~ +425 °C|
|(6) Process connection:||DN20/RF/14(or customized)|
|A: Al &ABS flap cylinder||B: Al flap plate|
|C: PVC ABS flap cylinder||D: PP&ABS flap cylinder|
|E: AL &LED||F: PP&LED|
|(8) Structure of top chamber:||1.Welding cover
2.welding cover with M14*1.5 plug screw
4.flange with M14*1.5 plug screw
5.welding cover with DN20 flange
6.flange with DN20 flange
7.welding cover with air valve
8.flange with air valve
|(9) Structure of bottom chamber:||1.flange cover
2.flange with drain plug screw
3.flange with DN20 welding tube
4.with DN20 flange
5.with needle/ball valve
|Application: Widely used in petroleum, chemical industry, oil field, pharmaceuticals industry, food, wine industry etc., suitable for level measurements under the environment of high/low temp., high pressure, strong corrosion, toxicity.|
|Note: Basic type, top mounted type, heat tracing type, LPG specialized type, anti-corrosive type, boiler steam specialized type etc, they have some differences in technical parameter. Different pressure has different flange specification.|
Sichuan VACORDA Instruments Manufacturing Co., Ltd.is a professional production-manufacturing enterprise specialized in designing, producing and marketing of instruments and apparatuses. Our company has been established for 23 years since 1997 to grow up to be a team of high-quality, united, hard-working and truth-seeking characters. Solid foundation has been established.
In 2017, we moved into our new factory whhich occupies the area about 20000 square meter, the workshop of 6900 square meter, R&D department of 1180 square meter, integrated office building of 1260 square meter, and we have established the high accuracy flow calibration lines.
Our company has got the ISO9001: 2000 international quality system certification in 2005, Import and export trading right in 2012, total 12 patents till 2016, the title of “national high-tech enterprise” and CE certificate for three categories product in 2014. Our company has owned trade marks as “vacorda”. At the same time, our company is preparing for more certificates for foreign market, such as IECEX, API, UL and so on, to lay a foundation for further international market.
Workshop and Skilled Workers
Commissioning before installing the magnetic level indicator with remote transmission
1. The resolution of Reed-11 dry reed transmitters is ±5mm and ±10mm. The user can select the type according to the actual needs of the measurement.
2. Before installation, debug the "0" bit of the remote transmitter and the "20mA" bit of the full-scale range. Put a piece of magnetic steel at the zero position of the sensor, make the lowermost reed switch close, adjust the "zero position" potentiometer, so that it outputs 4mA current. Then put the magnet in the upper limit position of the sensor, the uppermost reed pipe is closed, adjust the "range" potentiometer, so that it outputs 20mA current, and repeatedly debug several times until it is satisfied. After the debugging is completed, mark the place of "0" and the place of full scale.