Magnetic Level Gauge Price For Acid Alkaline Liquid Measuring
Widely used in petroleum, chemical industry, oil field, pharmaceuticals industry, food, wine industry etc. suitable for level measurements under the environment of high/low temp, high pressure, strong corrosion, toxicity.
|(1) Center-center distance:||150~6000mm|
|(2) Medium density:||More than 0.45g/cm3|
|(3) Nominal pressure:||PN2.5~PN40, PN40~PN320(x 0.1MPa)|
|(4) Material:||SS304, 316L, PP, PTFE, or customized|
|(5)Operating temperature:||-190 ~ +425 °C|
|(6) Process connection:||DN20~DN80/RF/14,DN80~DN250/RF/14|
|A: Al &ABS flap cylinder||B: Al flap plate|
|C: PVC ABS flap cylinder||D: PP&ABS flap cylinder|
|E: AL &LED||F: PP&LED|
|(8) Structure of top chamber:||
2.welding cover with M14*1.5 plug screw
4.flange with M14*1.5 plug screw
5.welding cover with DN20 flange
6.flange with DN20 flange
7.welding cover with air valve
8.flange with air valve
|(9) Structure of bottom chamber:||
2.flange with drain plug screw
3.flange with DN20 welding tube
4.with DN20 flange
5.with needle/ball valve
|Note: Basic type, top mounted type, heat tracing type, LPG specialized type, anti-corrosive type, boiler steam specialized type etc, they have some differences in technical parameter. Different pressure has different flange specification.|
1. Simple structure, good stability and reliability.
2. Independent of medium’s physical and chemical states such as conductivity, dielectric constants, foam etc.
3. Appliable for all kinds of medium environment such as corrosive, toxic and explosive one.
4. Interface measurement or level measurement of 2 kinds of medium with different density.
5. Two-wire 4~20mADC signal output, 0.8’’ or 0.56” LED digital display.
Workshop and Processing
The anti-corrosion type magnetic flap liquid level gauge is easy to show the phenomenon of false liquid level in long-term use. The reasons for forming the false liquid level are as follows:
|1.||First of all, make sure that the liquid phase and the gas phase are all connected|
|2.||Use a magnet to scan the whole process along its appearance, you will be able to understand that it will be corrected.|
|3.||If it is a safe liquid, first release the liquid phase and perform operation 2.|
|4.||If the vapor pressure of the gas phase is large, turn off the gas phase, press the liquid in the pipe back to the storage tank, and perform operation 2. The top-mounted magnetic flap liquid level gauge is suitable for measuring various containers that are not convenient for side-mounted liquid level gauges, especially the liquid level measurement in the public storage tank. It is generally applicable to industries and departments such as petroleum, chemical industry, metallurgy, electric power, light industry and medicine.|
|5.||The external interference signal is too large, the signal obtained by forming the circuit is a fake signal, not a signal measured in practice;|
|6.||The waveguide wire inside the liquid level gauge is defective, for example, it is loose, the seal is not good, the water is rusted, etc., which forms signal distortion;|
|7.||The waveguide device is incorrect and the signal transmission is distorted;|
|8.||The signal processing circuit is defective.|